Autosampler “HT2100H” (manufactured by “HTA s.r.l.” Italy-Brescia) is designed for automatic injection of equilibrium vapor samples (Headspace autosampler) into chromatographs of “Crystallux” series, and also “Crystal-2000”, LHM, model-3700, “Tsvet” series, etc.
RPC “Meta-chrom” is authorized distributor for products of “HTA s.r.l.” company in Russia. RPC “Meta-chrom” is authorized distributor for products of “HTA s.r.l.” company » in Russia».
Headspace sampler introduce volatile components from matrices into the chromatograph, the intorduction of which into the chromatograph is impossible or unwanted
Examples of such matrices are drinking and waste water, soil, biological fluids, food products and drinks, various polymers, building materials, pharmaceutical products, etc.
- Number of analyzed samples - from 1 to 14.
- Data exchange with PC - by standard interface LAN.
- Data exchange with chromatograph - by specific interface (TTL-logic).
- Vial volume for analyzes samples - 20 and 10 ml.
- Thermostat temperature for sample heating - from 40 to 150 °C.
- Volume of injected sample - from 0,01 to 5,0 ml (syringes 2,5, 5 and 1,0 ml), discreteness of sample injection - 0,01 ml.
- Possibility to inject the sample in series into two injectors.
- Standby time before and after injection - from 0 to 99 s.
- Sampling and injection speed - from 0,5 to 100 ml/min.
- Possibility of manual setting of the sampler for specific chromatograph by step-by-step programming from control panel, recording this setting parameters in memory, and also the depth of immersion of syringe needle into the vial.
- Setting any order of sample injection from vials on the tray.
- Syringe temperature (it is set independently) - from 40 to 150 °C.
- Number of syringe flushes - up to 15.
- Speed of shaking - 320-720 rpm.
- Shaking cycle (on/off) - from 0 to 10 min.
- Automatic transfer of vials with the sample to thermostat.
- Method of sample shaking - orbital.
- Possibility to clean the syringe by carrier gas purging.
- Number of injections of one sample - up to 15.
- Standby time between injections - from 0 to 100 min.
- Sample maintenance time in thermostat - from 0 to 999 min.
- Number of positions for sample heating in thermostat - 1.
- Delay for obtaining equilibrium during injection - from 0 to 10 min.
During work with chromatograph of “Crystallux” series sampler control is realized directly from chromatograph software without running additional programs.